Humidity, dew point and associated variables Variables used to describe humidity include : Partial vapor pressure Relative humidity Dew point Frost point Absolute humidity Mixing ratio Wet bulb temperature Parts per million Equilibrium relative humidity Water activity Enthalpy The expression moisture content mostly applies to solids and liquids. It is not a quantitative term when applied to a gas. Water vapor pressure (Pw) The pressure exerted by the water vapor present in air or a gas. Saturation vapor pressure of water (Pws) The maximum pressure the water vapor can exert at a particular temperature. The higher the temperature, the more water vapour the air can hold. HUMICAP (sensor) for relative humidity measurement Relative humidity (RH) This is the ratio of the partial water vapor pressure (Pw) to the water vapor saturation pressure (Pws) at a particular temperature. It is measured as a percentage: %RH = lOO%*(Pw/Pws) Relative humidity is strongly temperature dependent. Pressure will also change the RH. For example, if a process is kept at a constant temperature, the RH will increase by a factor of two if the pressure is doubled. Dew point temperature (Td) Is the temperature at which air becomes saturated with water and begins to condense, forming dew. At 100 %RH the ambient temperature equals the dewpoint temperature. The more negative the dewpoint is from the ambient temperature, the less the risk of condensation and the drier the air. Dewpoint is not temperature dependent, hut is affected by pressure. Frost point temperature (Tf) If the dewpoint temperature is below freezing the term frostpoint temperature is sometimes used. Tf is always slightly higher than Td below 0 °C as the water vapor saturation pressure of ice is different to water. Absolute humidity (a) Refers to the mass of water in a unit volume of moist air at a given temperature and pressure. Usually expressed as grams per cubic meter of air. It can be confused with mixing ratio. Mixing ratio (x) Is the ratio of the mass of water vapor to the mass of dry gas. It is dimensionless but often expressed in grams per kilogram of dry air. It is mainly used in drying processes or ventilation systems for calculating water content when the mass flow of air is known. Wet bulb temperature )Tw) Traditionally, this was the temperature indicated by a thermometer wrapped in a wet sheath. The wet bulb temperature and the ambient temperature can then he used to calculate relative humidity or dewpoint. Tables and charts can also be used. Parts per million (ppm) Is the ratio of water vapor to the dry gas or total (moist) gas. It is sometimes used for low levels of moisture and expressed by mass/volume or mass/weight. Equilibrium relative humidity (ERH) This is the value of the relative humidity of the air surrounding a substance when there is no net exchange of moisture. Water activity (aw) Is similar to equilibrium relative humidity and uses a scale of 0 to 1, instead of 0% to 100%. Enthalpy (h) Is the amount of energy required to bring a gas to its current state from a dry gas at 0 °C. Mostly used in air conditioning calculations. Vaisala has a choice of products for measuring relative humidity, temperature and dewpoint. Some products also have built-in calculation options to give outputs in terms of other humidity variables. The terms probe, transmitter and sensor are all used to describe a product that measures humidity. The probe is the part of the product that contains the humidity sensor and may he rigidly fixed to the transmitter housing or connected by a flexible cable. The transmitter provides the output signal. The critical part is the sensor. Vaisala designs and iiianufactures various families of sensors. HUMICAP - for relative humidity DRYCAP - for low dewpoint DEWCAP - for high accuracy dewpoint DEWCAP (sensor) for high accuracy measurement. What to look for in a sensor Range The majority of Vaisala RH sensors work over the full range from 0 to 100 %RH. The choice of product depends on the application and operating temperature especially at the extremes of humidity (>90 %RH and less than 10 %RH). Accuracy In general, this term is much abused and many unrealistic claims are made on data sheets. Vaisala products are compared and adjusted during manufacture against factory standards that are traceable to national standards. This calibration chain is detailed in the certificates now supplied with the majority of our products when shipped from the factory. Other important considerations associated with accuracy include: * Linearity over the working range * Hysteresis and repeatability * Stability over a period of time * Temperature dependence of the sensor Our aim is to provide "accuracy that lasts". This reflects our commitment to provide accurate and reliable sensors with excellent long-term stability. Response time Is the speed of response of the sensor when subjected to a step change in humidity. Other factors include temperature, airflow, and type of filter fitted. A blocked filter, for example, will give a slower response. Chemical tolerance Certain aggressive chemicals in the air can attack or contaminate the probe head or sensor. Vaisala sensors have good chemical tolerance and we have extensive application knowledge of the effect of various chemical concentrations. Examples of concentrations that may affect the sensor are: * Organic solvents - 1,000 to 10,000 ppm * Corrosive agents (strong acids) - 1 to 10 ppm * Weak acids - 100 to 1,000 ppm * Bases - 10,000 to 100,000 ppm Some products have a chemical purge (re-gain) feature. This heats the ensor and purges it of chemicals that may have gathered in the sensor over a period of time. Choice of product No single product can cover the full range of applications. Therefore the following must he taken into account:- Location Choose a measuring point that gives a true picture of the environment. Above 60 °C the transmitter electronics should be mounted outside the process and a suitable high temperature probe head inserted into environment. Beware of temperature differences When mounting the probe head into a process, avoid temperature drops along the body of the probe. If there is a large temperature difference between the probe head and external environment, the whole probe should be mounted in the process and the cable entry point insulated. If there is a risk of condensation, the probe should be mounted horizontally to avoid water dripping down the probe/cable and saturating the filter. See figure 1 Figure 1. Measurement error at 100%RH when the difference between the ambient and sensor temperature is 1 Deg.C. Air should he allowed to circulate around the probe head to ensure the sensor and process is at the same temperature. At 20 °C and 50 %RH a 1 °C difference between the sensor and zone of measurement will cause an error of 3 %RH. At 90 %RH the error is 6 %RH. See figure 2. Figure 2. Mounting arrangements. HUMITTER sensor Source: How to choose the right Vaisala product for measuring relative humidity and dewpoint. __________________________________________________ For details of Dew Point Sensors Suppliers, click here...

 Home - Website - Search - Suppliers - Links - New Products - Catalogues - Magazines Problem Page - Applications - How they work - Tech Tips - Training - Events - Jobs