Sensor Technology Ltd. - PRODUCT NEWS
Wireless torque sensor cuts out slip rings
The companys TorqSense Rotary Torque Sensors are unlike traditional slip ring transducers, using a simple non-contact radio link for collecting real-time torque signals. This means that machine builders and control engineers can do away with expensive and temperamental slip rings instead, simply mounting the sensors and checking the radio connection.
Electronics convert the data stream directly into engineering outputs, such as torque, speed, power and angle. Serial and USB interfaces make TorqSense directly compatible with PCs and other user-friendly features include built-in peak torque sampling, storage and torque averaging, and a self-diagnostics test package. The electronics can be provided, integral with the main unit or in a separate housing, to suit individual applications.
To achieve the non-contact operation that makes the TorqSense transducer range unique, SAW (surface acoustic waves) devices are used as frequency-dependent strain gauges to measure the change in resonant frequency, caused by strain experienced in the driveshaft. This measurement is directly related to the torque experienced in the rotating machine element.
In use, SAW devices are mounted onto the shaft and their output signals are transmitted by a low-power radio frequency (RF) coupled to an adjacent pick-up, which interprets the signals and converts them into a torque measurement readout or feeds them on into a PC for further calculation.
In a SAW sensor, the surface waves are produced by passing an alternating voltage across the terminals of two interleaved comb-shaped arrays, laid onto one end of a piezoelectric substrate. A receiving array at the other end of the transducer converts the wave into an electric signal.
The wave frequency is dependent upon the spacing of the teeth in the array and the direction of wave propagation is at right angles to the teeth. Therefore, any change in its length, caused by the dynamic forces of the shaft's rotation, alters the spacing of the teeth and hence the operating frequency. Tension in the transducer reduces the operating frequency while compression increases it.
To measure the torque in a rotating shaft, two SAW sensors are bonded to a shaft at 45° to the axis of rotation. When the shaft is subjected to torque, a signal is produced, which is transmitted to the adjacent stationary pick-up via the capacitive RF couple comprising two discs, one of which rotates with the shaft, the other being static.
The frequency of the oscillation used is typically 200MHz. The frequency-basis of the TorqSense concept gives a wide bandwidth and the susceptibility to electronic interference, common with other analogue-based techniques such as inductive devices, is eliminated.
About Sensor Technology
Sensor Technology Limited has been manufacturing quality torque measurement solutions for over 35 years and operates from two different locations in the heart of rural Oxfordshire; its sales, marketing, production and calibration facilities located at Apollo Park, Wroxton and a research and development facilities at Balscott Mill, Balscote. Sensor Technology also has a global network of distributors assisting sales and a network of world-renowned technical experts assisting the development of TorqSense and other technologies.
For more information, please contact :-
Home - Website - Search - Suppliers - Links - New Products - Catalogues - Magazines Problem Page - Applications - How they work - Tech Tips - Training - Events